Home < HM System < Prosthetic Procedure < Denture
HM Implant Components
Precaution & Procedures
Maintenance & Sterilization
Insert black O-ring in the retainer, place the retainer on the ball head exposed in the mouth and take an impression.
Connect lab analog to the retainer in the impression.
After manufacturing the working model, remove the retainer from the impression according to the general rules.
Using the completed working model, manufacture denture according to the general rules.
Check the insertion points of the denture.
Push the O-ring through the rubber dam. Insert the yellow O-ring into the retainer. Attach the retainer to the O-ring abutment.
Inject resin into the hole formed on denture and remove the surplus after resin has hardened.
Check the bearing and occlusion status.
Considering path, insert HM implant according to the set surgery procedure.
After checking the location of insertion, form the hole on the existing denture.
Connect rubber dam and then connect ball head by inserting O-Ring with weak bearing into the retainer.
Inject resin after setting the existing denture formed hole. Remove the surplus part after resin has hardened.
Check the retention force and occlusion status.
HG LOCATOR Abutment